EB is based upon an exhaled breath sample matrix. A passive collection device containing a thin electret (electrostatic-based) filter is used to capture submicron bioaerosol particles originating from airway lining fluid of the lungs. The single barrel collection device is a one-time-use, low cost and compact unit. Each device is supplied individually boxed in a package containing everything needed for one sample collection procedure, including clear diagrammatic-based user instructions. The procedure is quick and easy for both the donor and collector.
See user instructions. (sensabues-user-instructions)
EB has been developed into a platform that offers users a complete solution to collect, extract, detect and identify non-volatile compounds present in exhaled breath. This is a back-to-lab based testing methodology for specimen collection and detection using the analytical technique of high performance liquid chromatography-tandem mass spectrometry (HPLC-MS/MS).
As an integral part of the functional design, a small fraction of the exhaled breath passing through the collection device is diverted into a compact clear plastic bag. This bag serves two functions. Firstly, by inflating, it is a visual indicator that the donor is exhaling breath through the device correctly. Secondly, when fully inflated, it indicates that the minimum required amount of air (~30 litres) has passed through the filter, which equates to ~3 minutes of normal tidal breathing, or less if the donor prefers to force air through the device.
Sensabues has applied for, and been granted, patents for key design features and aspects of the EB sampling device, system and method.
More information on this technology and product is available to download from the RESOURCES page on this website, including a list of the published technical papers and a list of FAQs.